|Capacity||up to 100 mm sample height|
|Temperature||up to 100degC|
The tests are performed in a triaxial cell capable of supporting up to 100 MPa for confining and deviatoric stress on geo-materials cored or manufactured in diameters ranging from 20 mm to 100 mm and heights from 40 to 200 mm. Multiple sensors can continuously monitor axial and radial deformations of the sample, as the parameters of pressure applied. The tests can be performed at room temperature, but there are also different versions of the equipment that can be heated up to 200 °C.
In addition to the axial and radial pressures that can carry many mechanical tests, the sample can be subjected to a pressure drainage, gas or liquid. It is then possible to perform permeability tests.
The unit is managed with an Industrial Control System that allows real time monitoring all measured parameters and controlling the different pressure circuits.
Self Compensated Cell
An intermediate chamber compensates the axial stress created by the confining pressure on the sample and changes in confining pressure have no influence on the axial pressure.
A Full potential of Testing
This unit may nevertheless be used for axisymmetric testing by sealing the chamber circuit.
The first tests presented below are used to measure the mechanical properties of a sample.These tests use independently or simultaneously, the opportunities provided by the axial pressure and the confining pressure. Each circuit for axial or radial stress is controlled by a High Pressure Metering Pump to a level of constant, variable, or cyclic pressure. The circuits are controlled by an industrial control system. All thesetests can be performed without drainage in the sample or a sample put under pressure by internal drainage.
The second series of tests presented to measure the permeability of a sample while subjecting it to fixed or variable mechanical stresses. As for mechanical tests, permeability tests can be performed at room temperature but also at temperatures up to 200 ° C.
Finally, it is also possible to use the machine as a simple Rock Stac press by removing the containment. It is thus possible to perform standard tests of fracture mechanics without drainageand room temperature.